DRYSDALE POLAR A.C.POTENTIOMETER PDF
A.C. POTENTIOMETER. Its principle is same as a But in an a.c. potentiometer, not only the magnitudes but phases as well DRYSDALE POLAR TYPE AC. Posts about polar type ac potentiometer written by Pragnarongali. The working principle of the AC potentiometer is same as that of the DC potentiometer, i.e., the unknown voltage is determined by Polar Type Potentiometer.
|Published (Last):||18 July 2005|
|PDF File Size:||12.37 Mb|
|ePub File Size:||11.59 Mb|
|Price:||Free* [*Free Regsitration Required]|
This principle of operation of all potentiometers is based on the following basic circuits.
The working current is standardized by reference to a known voltage source as a first a.d.potentiometer in the measurement. But there is one major difference between their measurements, DC potentiometer only measures the magnitude of the unknown voltage. Now to measure unknown e. Precision laboratory potentiometers usually have three voltage ranges a high range 1. The protective resistance is left in the circuit to avoid damage to the galvanometer during the initial stages of adjustment.
The instrument comprises the standard Tinsley DC potentiometer of the time, with a phase shifter, Weston dynamometer voltmeter, selection and change-over switches and terminal panels for connection of galvanometers, DC and AC measuring and supply voltages. Arrangements are provided for reversing the ammeter in order crysdale eliminate the effect of stray fields. It gives the accurate results in magnetic testing and precise testing of instrument transformers. This working current through the slide wire may be varied by changing the rheostat setting.
The main circuit of the potentiometer must be supplied with alternating current of exactly the same frequency as that in the circuit drjsdale be measured, which means in practice that they must be supplied from the same source. In a phase-shifting transformer, there is a combination of two ring-shaped laminated steel stators connected perpendicularly to each other as shown in the figure. The method of measuring unknown voltage E depends on finding a position for a.c.potentiomfter sliding contact such that the galvanometer shows zero deflection when galvanometer key K is closed.
The whole arrangement of the winding is done in such a way that the magnitude of the induced emf in the rotor may change but it does not affect the phase angle and it can be read on the scale fixed on the top of the instrument.
The potentiometer – is then thrown on to the A. There is a provision on the scale that it could read phase angle up to The indicator for low frequencies is a vibration galvanometer which must be closely tuned to resonate to the frequency of the circuit. The phase shifting transformer is connected to the potentiometer to vary the phase of the voltage on the current. The main potentiometer current must be kept at some known constant value.
The potentiometer method is the usual basis for the calibration of voltmeters, ammeters and watt meters. The potentiometer, which is the universal instrument by which currents are determined in terms of a difference of potential between the end of a known resistance, has been developed by C.
There are two types of AC potentiometers: Thus, the resultant voltage of the coordinate AC potentiometer is And the phase angle is given by.
Leave a Reply Cancel reply Your email address will not be published. The working current is checked by returning to the calibrate position.
The instrument is balanced a.c.poteentiometer the main dial and the slide wire, leaving the protective resistance again in the circuit.
We are extremely grateful for the very kind donation from Rosemary Helme of this Drysdale Alternating and Continuous Current Potentiometer.
Between the stators, there is laminated rotor having slots and winding which supplies voltage to the slide-wire circuit of the Potentiometer. Coordinate type Potentiometer In coordinate AC potentiometertwo separate potentiometers are caged in one circuit as shown in the figure.
This ammeter must read correctly with alternating and continuous currents. The combination of dial resistors and slide wire is set to the value of the standard cell voltage. Drysdale for the measurement of alternating potentials. Measures voltage as magnitude and phase from reference voltage.
It is an instrument srysdale to measure an unknown voltage by comparing it with a known voltage. As it is necessary to have the splitphase circuit somewhere near the resonating point, it must be adjusted for changes of frequency. In this potentiometer, the magnitude of the unknown voltage is measured from one scale, and its phase angle is directly on the second scale.
There are two step-down transformers T 1 and T 2 which isolate potentiometer from the line and give an earthed screens protection between the winding.